|Materials:||ABS Plastic And Electronic Components||Color:||White And Green / Customized|
|Measuring Distance:||2-5cm||Power:||Removable Batteries DC3V|
|Battery:||Built In 2*AAA Alkaline Dry Bettery||Product Dimension:||116*55*170m|
|Display:||Digital LCD Screen||IP Rating:||IP 20|
|Production Validity:||3 Years||Temperature Mode:||Body Mode,object Mode|
|Measuring Range:||Body:,89~109°F/32~42°C Object:32~212°F/0~100°C||Instrument Classification:||Class II|
|Safety Standard:||GB 9706.1-2007||Warranty:||1 Year|
|Accuracy:||+/-32°F And +/-0.2℃,|
RoHs Medical Infrared Thermometer,
Buzzer 30DB digital ir thermometer,
150g ear and forehead thermometer
150g Ear And Forehead Object Surface Medical Infrared Thermometer FDA RoHs
Kids Adults Object Surface Non Contact Infrared Forehead Digital Thermometer With FDA RoHs
|Plastic ABS and electronic component|
2*1.5V AAA Batteries
Buzzer, 30 DB , Switch control
|Measuring range||Body:,89~109°F/32~42°C Object:32~212°F/0~100°C|
|Operating ambient temperature||50~104°F/10~40°C|
|Operating ambient humidity||Less than or equal to 80%RH|
|Storage temperature range||-13~-131°F/-25~-55°C|
Net Weight:150g(without batteries)
Carton Size:605*360*295mm 50pcs/carton
1. Please read the instructions for use before the measurement, and the forehead
should be kept dry, and the hair should not cover the forehead (please perform the
measurement in an environment of 10℃-40℃) to ensure the accuracy of the measurement.
2. The forehead temperature measured quickly by this product is for reference only
and should not be used as a basis for medical judgment. If any abnormal body temperature
is found, please use a medical thermometer for further measurement.
3. Please protect the sensing lens and clean it in time. If the changing environment
temperature changes too much, you need to place the measuring instrument in the
environment you want to measure for 20 minutes, and wait for it to stabilize to adapt
to the environment temperature before using it to get a more accurate value.
1.Switch between body and object temperature mode
when in measurement mode,press the “SET”key to switch between the object
temperature measurement mode(the display shows a small house) and the human
body temperature measurement mode(the display shows the child's smiling face)
In the shutdown state,press the “MEMO” key to view the last first measured data,and
press it again to view the last second data.The thermometer can store 32 sets of data
measured in the human body temperature measurement mode.Press the “POW”key
to exit the look up.
In the shutdown state,press and hold the “SET”key to 3 seconds until the screen
lights up.in F1 mode,press “MEMO” to select on or off.Press “POW” to exit the setting.
In F1 mode,lightly press “SET” button to enter F2 mode,in this state,press “MEMO”
button to set the temperature.Press “POW” to exit the setting.
In F2 mode,lightly press “SET” button to enter F3 mode,
press press “MEMO” button to set the temperature compensation.This operation is
more professional,it is recommended to keep the factory default settings and set the
value to zero.Press “POW” to exit the setting.
In F3 mode,lightly press “SET” button to enter F4 mode,press
“MEMO” key to set the temperature unit(Celsius/Fahrenheit switching selection).
Q1:The device is on (the screen is measured),but the temperature cannot be measured.
A1:Confirm whether the product enters the measurement data viewing mode or parameter
setting mode.If you under this mode,please press the power button(POW key) to shut down
(or wait 1 minute to automatically shut down),and then press the temperature measurement
key after rebooting.
Q2:The temperature measurement is obviously too low
A2:Confirm whether the device is under the object temperature measurement mode.If so,
press the SET key once to enter the human body temperature measurement mode.Enter
the parameter setting mode and check whether the temperature compensation setting is
negative.It is recommended to set the value to 0.
Q3:The measurement temperature is obviously too high
A3:Enter the parameter setting mode and check whether the temperature compensation
setting is positive.It is recommended to set the value to 0.
Contact Person: Mary Yang