Fingertip Pulse Oximeter
1.A Battery life indicator
2.In the absence of signals , the product would power-off automatically after 8 seconds
3.Small,light in weight, and convenient to carry
Keys function description: In standby mode, start the key instrument into the working state, Touch this button under the working state can change the display mode.
Product Basic Parameters
Display Type: OLED display
Accuracy:±2% on the stage of 80%~99%
±3% on the stage of 70%~79%
Below 70% no requirement
PR: Measurement range:330BPM~240BPM
Accuracy:±1BPM or ±1%(the larger one)
Power: two AAA 1.5V alkaline batteries
Power consumption: below 30mA
Automatic power-off: The product power off by itself when no finger is in the product for≥8 seconds
Transport, storage environment:
Atmospheric pressure: 50kPa ~ 106kPa,non-corrosive gas and well-ventilated environment
1.Install a pair of AAA batteries into the battery compartment.
2.Plug a finger into a rubber hole of the Oximter.
3.Press button on the front panel to start.
4.Do not shake your finger or move your body when the Oximeter is working
5.Press the button if you want to change display direction
6.Read the relevant value from the display screen
7.The device will automatically power off in 8 seconds if there is no finger signal.
Carton Total Weight: ≈ 5.5kg(Gross Weight)
PCS/Carton : 100 PCS
Carton Size: 475mm*210mm*275mm, ≈0.028CBM