150g non contact temperature gun,
3V non contact temperature gun
|Product Name||Baby Forehead Infrared Non Contact Fever Alarm Accurate Instant Reading Thermometer|
|Response Time||1 second|
|Memory||32 memory in 1 group|
|Display Model||Digital LCD|
|Unit weight||150g (without batteries)|
|Battery||3V; 2*AA battery|
Forehead temperature gun
1. Human body temperature measurement: accurate measurement of human body temperature,
replacing traditional mercury thermometers. Women who are ready to have children can use infrared thermometers (forehead thermometers) to monitor basal body temperature, record the body
temperature during ovulation, and choose the right time to conceive, and can also measure the
temperature to judge pregnancy and so on.
Of course, the most important thing is to observe whether there is any abnormality in your body
temperature at any time, to avoid influenza, and to prevent swine flu.
2. Skin temperature measurement: to measure the surface temperature of the human skin, for example,
it can be used for re-implantation of a limb to measure the surface temperature of the skin.
3. Object temperature measurement: measure the surface temperature of an object, for example,
it can be used to measure the surface temperature of a teacup.
4. Liquid temperature measurement: measure the temperature of the liquid, such as the temperature
of the baby's bath water. When the baby takes a bath, measure the water temperature, no longer worry
about cold or hot; it can also measure the water temperature of the milk bottle to facilitate the preparation
of Baby milk powder
How to use it?
1.Switch between body and object temperature mode
when in measurement mode,press the “SET”key to switch between the object temperature measurement mode(the display shows a small house) and the human body temperature measurement mode(the display shows the child's smiling face)
In the shutdown state,press the “MEMO” key to view the last first measured data,and press it again to view the last second data.The thermometer can store 32 sets of data measured in the human body temperature measurement mode.Press the “POW”key to exit the look up.
In the shutdown state,press and hold the “SET”key to 3 seconds until the screen lights up.in F1 mode,press “MEMO” to select on or off.Press “POW” to exit the setting.
In F1 mode,lightly press “SET” button to enter F2 mode,in this state,press “MEMO” button to set the temperature.Press “POW” to exit the setting.
In F2 mode,lightly press “SET” button to enter F3 mode,
press press “MEMO” button to set the temperature compensation.This operation is more professional,it is recommended to keep the factory default settings and set the value to zero.Press “POW” to exit the setting.
In F3 mode,lightly press “SET” button to enter F4 mode,press
“MEMO” key to set the temperature unit(Celsius/Fahrenheit switching selection).
Q1:The device is on (the screen is measured),but the temperature cannot be measured.
A1:Confirm whether the product enters the measurement data viewing mode or parameter setting mode.If you under this mode,please press the power button(POW key) to shut down(or wait 1 minute to automatically shut down),and then press the temperature measurement key after rebooting.
Q2:The temperature measurement is obviously too low
A2:Confirm whether the device is under the object temperature measurement mode.If so,press the SET key once to enter the human body temperature measurement mode.Enter the parameter setting mode and check whether the temperature compensation setting is negative.It is recommended to set the value to 0.
Q3:The measurement temperature is obviously too high
A3:Enter the parameter setting mode and check whether the temperature compensation setting is positive.It is recommended to set the value to 0.
Contact Person: Mary Yang