|Materials:||ABS Plastic And Electronic Components||Color:||White And Green|
|Battery:||Built In 2*AAA Alkaline Dry Bettery||Product Dimension:||116*55*170m|
|Display:||Digital LCD||IP Rating:||IP 20|
|Production Validity:||3 Years||Temperature Mode:||Body Mode,object Mode|
|Measuring Range:||Body:,89~109°F/32~42°C Object:32~212°F/0~100°C||Instrument Classification:||Class II|
|Safety Standard:||GB 9706.1-2007||Warranty:||1 Year|
Digital Infrared Thermometer Buzzer 30DB,
150g non contact temperature gun,
GB 9706.1-2007 Infrared Thermometer
High Accuracy Forehead Digital Infrared Thermometer Buzzer 30DB
1, One second measurement, simple and convenient;
2, 32 sets of memory data;
3, Large screen backlight display;
4, Safe and practical, long service life;
5, Three colors change, the fever will show red
6, Advanced probe, accurate measurement;
7, Temperature and body temperature measurement one-click conversion;
8, Celsius and Fahrenheit convertible with adjustable accuracy
|Poduct Name||High Accuracy Non Contact Forehead Infrared Thermometer Best Price|
|Plastic ABS and electronic component|
CE FDA ROHS ISO13485
2*1.5V AAA Batteries
Buzzer, 30 DB , Switch control
|Measuring range||Body:,89~109°F/32~42°C Object:32~212°F/0~100°C|
|Operating ambient temperature||50~104°F/10~40°C|
|Operating ambient humidity||Less than or equal to 80%RH|
|Storage temperature range||-13~-131°F/-25~-55°C|
How to use it?
1.Switch between body and object temperature mode
when in measurement mode,press the “SET”key to switch between the object
temperature measurement mode(the display shows a small house) and the human
body temperature measurement mode(the display shows the child's smiling face)
In the shutdown state,press the “MEMO” key to view the last first measured data,
and press it again to view the last second data.The thermometer can store 32 sets
of data measured in the human body temperature measurement mode.Press the
“POW”key to exit the look up.
In the shutdown state,press and hold the “SET”key to 3 seconds until the screen
lights up.in F1 mode,press “MEMO” to select on or off.Press “POW” to exit the setting.
In F1 mode,lightly press “SET” button to enter F2 mode,in this state,press “MEMO”
button to set the temperature.Press “POW” to exit the setting.
In F2 mode,lightly press “SET” button to enter F3 mode,
press press “MEMO” button to set the temperature compensation.This operation is
more professional,it is recommended to keep the factory default settings and set the
value to zero.Press “POW” to exit the setting.
In F3 mode,lightly press “SET” button to enter F4 mode,press
“MEMO” key to set the temperature unit(Celsius/Fahrenheit switching selection).
Who we are?
Hunan Zhongliankang Medical TechnologyCo.Ltd. is localed in Xiangtan Hunan province.
The main business of our enterprise is home-used medical equipment like infrared forehead
thermomrter . Our workplace and production and inspection equipment are determined to
meet the-characteristics and processing requirements of the thermometer.
Our major customers include chain pharmacies and individual pharmacy, rehabilitation
institutions and medical institutions. We also provide logistics, and other value-added
services to meet the needs of our customer and the "one stop shopping" service for business.
Q1:The device is on (the screen is measured),but the temperature cannot be measured.
A1:Confirm whether the product enters the measurement data viewing mode or parameter
setting mode.If you under this mode,please press the power button(POW key) to shut down
(or wait 1 minute to automatically shut down),and then press the temperature measurement
key after rebooting.
Q2:The temperature measurement is obviously too low
A2:Confirm whether the device is under the object temperature measurement mode.If so,
press the SET key once to enter the human body temperature measurement mode.Enter
the parameter setting mode and check whether the temperature compensation setting is
negative.It is recommended to set the value to 0.
Q3:The measurement temperature is obviously too high
A3:Enter the parameter setting mode and check whether the temperature compensation
setting is positive.It is recommended to set the value to 0.
Contact Person: Mary Yang