|Color:||White And Green / Customized||Materials:||ABS Plastic And Electronic Components|
|Measuring Distance:||2-5cm(recommend)||Power:||Removable Batteries|
|Battery:||Built In 2*AAA Alkaline Dry Bettery||Product Dimension:||116*55*170m|
|Weight:||150g||Sound:||Buzzer 30DB Swift Contral|
|Display:||LCD Screen||IP Rating:||IP 20|
|Temperature Mode:||Body Mode,object Mode||Measuring Range:||Body:,89.6~109.6°F/32~42°C Object:32~212°F/0~100°C|
|Instrument Classification:||Class II||Production Validity:||3 Years|
|Warranty:||1 Year||Accuracy:||Plus Or Minus 0.2℃/32°F|
|Hot Spots:||Ovulation Test For Women To Prepare For Pregnancy/baby/child/Adults/Old Man|
Care4U Temperature Gun,
baby thermometer forehead and ear
Body Object Mode Digital Non Contact Infrared Thermometer
Product basic parameters
Measuring distance:2cm-5m(recommend measurement distance)
Sound:Buzzer, 30 DB , Switch control
Temperature measurement range:Body:,89.6~107.6°F/32~42°C Object:32~212°F/0~100°C
mode can be swift
Storage:Store temperature data up to 32 data sets
Battery:2*1.5V AAA Batteries
Low battery consumption:automatic shutdown without operation for 60 seconds
Net Weight:150g(without batteries)
Carton Size:605*360*295mm 50pcs/carton
HD LCD Screen
Non Contact Measuring
One Second Reading
Body and Surface Mode
32 Record Storage
1. Please read the instructions for use before the measurement, and the forehead should be kept dry, and the hair should not cover the forehead (please perform the measurement in an environment of 10℃-40℃) to ensure the accuracy of the measurement.
2. The forehead temperature measured quickly by this product is for reference only and should not be used as a basis for medical judgment. If any abnormal body temperature is found, please use a medical thermometer for further measurement.
3. Please protect the sensing lens and clean it in time. If the changing environment temperature changes too much, you need to place the measuring instrument in the environment you want to measure for 20 minutes, and wait for it to stabilize to adapt to the environment temperature before using it to get a more accurate value.
1.Switch between body and object temperature mode
when in measurement mode,press the “SET”key to switch between the object temperature measurement mode(the display shows a small house) and the human body temperature measurement mode(the display shows the child's smiling face)
In the shutdown state,press the “MEMO” key to view the last first measured data,and press it again to view the last second data.The thermometer can store 32 sets of data measured in the human body temperature measurement mode.Press the “POW”key to exit the look up.
In the shutdown state,press and hold the “SET”key to 3 seconds until the screen lights up.in F1 mode,press “MEMO” to select on or off.Press “POW” to exit the setting.
In F1 mode,lightly press “SET” button to enter F2 mode,in this state,press “MEMO” button to set the temperature.Press “POW” to exit the setting.
In F2 mode,lightly press “SET” button to enter F3 mode,
press press “MEMO” button to set the temperature compensation.This operation is more professional,it is recommended to keep the factory default settings and set the value to zero.Press “POW” to exit the setting.
In F3 mode,lightly press “SET” button to enter F4 mode,press
“MEMO” key to set the temperature unit(Celsius/Fahrenheit switching selection).
Q1:The device is on (the screen is measured),but the temperature cannot be measured.
A1:Confirm whether the product enters the measurement data viewing mode or parameter setting mode.If you under this mode,please press the power button(POW key) to shut down(or wait 1 minute to automatically shut down),and then press the temperature measurement key after rebooting.
Q2:The temperature measurement is obviously too low
A2:Confirm whether the device is under the object temperature measurement mode.If so,press the SET key once to enter the human body temperature measurement mode.Enter the parameter setting mode and check whether the temperature compensation setting is negative.It is recommended to set the value to 0.
Q3:The measurement temperature is obviously too high
A3:Enter the parameter setting mode and check whether the temperature compensation setting is positive.It is recommended to set the value to 0.
Contact Person: Mary Yang