|Temperature:||10℃~40℃||SpO2 Measuring Range:||0%- 100%,|
|Pulse Rate Measuring Range:||30 Bpm - 250 Bpm,||Power Requirements:||2 X1.5V AAA Alkaline Battery|
|Power Consumption:||Smaller Than 30mA||Resolution:||1% For SpO2 And 1 Bpm For Pulse Rate.|
|Measurement Accuracy:||±2% In Stage Of 70%-100% SpO2||Auto Off:||8 Seconds.|
2 AAA Batteries Blood Oxygen Tester,
ISO13485 Blood Oxygen Tester
Fingertip Blood Oxygen Tester Oximeter Monitor and Heart Rate Tester-Easy to Use, Small in Volume, Low Power Consumption
➢Power consumption of the product is low and the two originally equipped AAA batteries can be operated continuously for 20 hours.
➢The product will automatically be powered off when no signal is in the product within 5 seconds.
Small and portable the size of a matchbox, easy to carry, anytime, anywhere, just test
High precision one-key measurement, just press the key to start the measurement.
Four directions display the reading is more comfortable
Automatically detect shutdown save energy
Major Applications and Scope of Application
The pulse oxygen saturation is the percentage of HbO2 in the total Hb in the blood, so-called the O2 concentration in the blood. It is an important bio-parameter for the respiration .For the purpose of measuring the SpO2 more easily and accurately, our company developed the Pulse Oximeter.
The Pulse Oximeter can be used to measure human Hemoglobin Saturation and pulse rate through finger, and indicate the pulse intensity by the bar-display. The product is suitable for use in family, hospital (Ordinary sickroom), Oxygen Bar, social medical organizations and also the measure of saturation oxygen and pulse rate.
|Display Information||Display Mode|
|The Pulse Oxygen Saturation(SpO2)||LED|
|Pulse Intensity (bar -graph)||LED|
|SpO2 Parameter Specification|
|Measuring range||0%~-100%, (the resolution is 1%).|
|Accuracy||70%~100%±2%, Below 70% unspecified.|
Red light (wavelength is 660nm)
Infrared (wavelength is 880nm)
|Pulse Parameter Specification|
|Measuring range||30bpm~250bpm (the resolution is 1 bpm)|
|Accuracy||±2bpm or±2% select larger|
|Range||Continuous bar-graph display, the higher display indicate the strong pulse|
|1.5V (AAA size) alkaline batteries x 2 or rechargeable battery|
|Battery Useful Life|
|Two batteries can work continually for 20 hours|
|Dimensions and Weight|
|Dimensions||57(L) x 31(W) x 32(H) mm .|
|Weight||About 50g (with the batteries)|
1.Install a pair of AAA batteries into the battery compartment.
2.Plug a finger into a rubber hole of the Oximter.
3.Press button on the front panel to start.
4.Do not shake your finger or move your body when the Oximeter is working
5.Press the button if you want to change display direction
6.The device will automatically power off in 8 seconds if there is no finger signal.
Contact Person: Albert Li